LSP Oxygen Regulator Recall
The National Institute for Occupational Safety and Health (NIOSH) has determined that aluminum body regulators used in high-pressure oxygen systems are at risk for fires and explosions. The NIOSH report recommends immediately replacing all aluminum regulators used in high-pressure oxygen systems with brass regulators. Flash fires can result from friction and heat generated by particle impact during the release of high-pressure oxygen flowing through the aluminum regulator. The fires can start in aluminum oxygen regulators at pressures as low as 25 pounds per square inch (psi), while brass regulators will not ignite at pressures below 10,000 psi.
At the IAFF’s request, NIOSH initiated an investigation in July 1998 to review the oxygen regulator fire that resulted in severe burns to a member of IAFF Local 3333 Broward County, FL. The NIOSH investigation also included technical experts from the FDA, the National Aeronautic and Space Administration (NASA), as well as additional outside experts. There have also been oxygen regulator fires in numerous IAFF locals: Austin, TX Local 975; Houston, TX Local 341; Greeley, CO IAFF Local 888.; Springfield, OR IAFF Local 1395; Gurnee, IL IAFF Local 3598; and most recently in Oak Creek, WI Local 1848; and Truckee Meadows, NV Local 2487. The FDA is aware of at least 16 explosions in the past five years. Every aluminum regulator used on high-pressure oxygen cylinders has the potential to ignite a flash fire.
Prompting the Investigation
Soon after the IAFF heard of a similar incident in Austin, Texas, the IAFF again contacted the FDA and requested an investigation. Again, the IAFF was told that an investigation had to be initiated by the device’s owner. The IAFF asked if similar incidents had occurred, and was informed that they do not provide this information. In April 1997, the IAFF requested copies of all reports, MedWatch complaints, FDA directives, inspections, and any other information related to Life Safety Products' (LSP) or any other manufacturer's oxygen regulators utilized on emergency resuscitation oxygen tanks from January 1, 1995. This request was made through the Freedom of Information Act.
The IAFF finally received the information from FDA, but all the information was on microfiche--the only way they claimed they could provide this information. The IAFF had to have all the film then printed. A review of the information revealed that most, if not all of the 1,000s of pages failed to any useful information. The information was essentially page after page of computer printout of complaints about resuscitators. The files did not contain investigative reports or recall notices. The printouts also involved all oxygen resuscitator companies, not just LSP.
In May 1997, Allied Healthcare Products issued a "recall" of oxygen regulator model L270-020 and L270-050. The "recall" was really a notice to retrofit existing aluminum regulators with a sintered brass mesh screen to filter out contaminants.
Getting the Word Out
On February 4, 1999, Allied Healthcare Products issued a new "recall" to replace its LSP aluminum oxygen regulators with brass regulators. This is a voluntary action by the manufacturer and not mandated by FDA. Furthermore, the "recall" only affects LSP regulators, approximately 60 percent of the high-pressure oxygen regulator market. The FDA has done nothing to ensure that the remaining manufacturer’s regulators will be recalled.
In light of the NIOSH report, the IAFF immediately sent a letter to the Food and Drug Administration (FDA) demanding a full recall of all aluminum body regulators used in high-pressure oxygen cylinders. Furthermore, the IAFF demanded that the FDA require each manufacturer to contact each regulator owner, document the contact, and provide the retrofit brass regulators at no cost to the owners.
IAFF members should immediately check the high-pressure oxygen regulators owned by their fire department. All aluminum regulators must be replaced with brass regulators. Until the aluminum regulators are replaced, the NIOSH and FDA report has made interim recommendations.
Allied Healthcare Products, manufacturer of LSP regulators, has agreed to replace the aluminum regulators with brass regulators. If a different company manufactures the aluminum regulators, fire departments should contact that company or the distributor from whom they bought the regulator. Fire departments should demand that the aluminum regulators be replaced with brass. Fire Departments can contact the Allied Healthcare Products recall coordinator at:
LSP Regulator Recall Center
For more information please contact the IAFF’s Department of Occupational Health and Safety.
What You Can Do
Oxygen Regulator Use
Additional Links of Interest